Discuss.get support. exchange your experience.
Posted: 28.01.2021 01:32
Modified: 03.12.2023 01:22
It can be changed from the Aranet4 app as shown in the picture below:
Posted: 09.10.2021 08:33
The measurement interval won't change on my recently purchased Aranet4. An error message appears: Failed to change measurement interval.
Posted: 11.10.2021 09:12
Measurement interval can't be changed from Aranet4 app if the sensor is paired to Aranet Pro base. I guess it's not your case?
Please check and upgrade if needed the firmware version of your Aranet4 sensor: https://forum.aranet.com/all-about-aranet4/how-can-i-update-aranet4-device-firmware/
Reinstall of Aranet4 app and power restart of the sensor can help. If not, please contact firstname.lastname@example.org.
Posted: 01.10.2022 19:21
I have an Aranet4 Pro with firmware 0.4.10 and *not* paired with a base station yet the measurement interval is solid black in the app and cannot be changed. How do I change it?
Posted: 03.10.2022 08:07
Please move Radio switch to Off (lower) position and try changing interval again.
You have to log in to reply.